Call: +44 20 3289 8840
Dental Instruments Category
This Autoclave is specially designed for the occasions in use of frequent sterilizations, which can be applied in sterilizing for the tooth device, operation device, Sanitary Material and other heat-resisting articles existing in the Dept. of Stomatology, Ophthalmology and lab. It is equipped with the microprocessor for intelligent control and man-machine interface for convenient operation. With the distinctions in displaying parameter and condition actively, evaluating malfunction automatically, shutting off for self-protection automatically in case of excessive temperature and pressure.
1. Fuzzily computer controlled and digital bright show. The interface is modularized panel for easy operation.
2. The alarm system of full waste water completely avoids the waste water circulation sterilizing.
3. With automatic alarm system of breakdowns, you can grasp the status of machine operation dynamically at any time.
4. Humanized design of water drain makes it more convenient to dry.
5. With built-in printer, the process of sterilization will be conveniently recorded.
6. Double-layer protection：Door lock with interlock protection, inter-cavity pressure, the door cannot be opened.7. Sound hint：Buzzing hints when disinfection is over or system is lack of water.8. Convenient maintenance aided by auto failure detection.
|Rated Power||1800VA / 2000VA|
|Sterilizing (Pressure/temp.)||0.9-1.3bar/120ºC, 2.0-2.3bar/120ºC (Both of above are Under Standard Atmospheric pressure)||Fuse-Melting pipe||T15A / T30A|
|Capacity of water tank||2L||Water Consumption volume per recycling round||0.3L|
|Operating Temp||0-40ºC||Inside dimensions||245*355mm|
|Outside dimensions||650*520*500 mm||Weight||56kg|
1. The autoclave must be placed on the horizontal working base.
2. Must use the distilled water for the purpose of lasting the working life.
3. There shall be no clog to the air cooling window on the outer surface.
4. The object apparatus to be sterilized shall be placed on the device plate with enough clearance retained in favor of the air circulation inside the sterilizing room.
5. Empty the water inside the cooling water collecting tank frequently. Usually, the cooling water collecting tank shall be emptied once the water storing tank emptied.
6. Must revolve the door handle to the terminal side while working.
7. Don't open the door of autoclave when seeing the "0" displayed on the pressure indicator.
8. Not be too closed to the door of the autoclave in case of scald when open the door.
9. Shut off the power before discharging/installing the sealing ring, and the work shall be carried out after further sufficient cooling in case of scald.
10. Don't drag or drop the autoclave during relocation period of the autoclave.
11. The protecting ground shall be reliable.12. Equipment must be used away from the magnetic field.
|Product Name||Quantity||Product Name||Quantity|
|Autoclave Sterilizer Vacuum Steam with printer||1||Draining pipe||1|
|Measuring cup||1||Sealing ring||1|
|O-Ring of tap||2||Printer Paper||1|
1. Orders will be shipped to verified address, please make sure it’s correct before paid.
All shipping address needs PHONE NUMBER.
2. 98% items will be dispatched within 1-3 working days after your payment confirmed.
3. Our paypal account is firstname.lastname@example.org. Money sent is available.
1. Visa, Master Card, Debit are all ok for payment.
2. Credit Card on Paypal is ok.
1. Please contact email@example.com to get more details.
1. Please contact firstname.lastname@example.org to get bank account.
1. ZetaDental.com offers global free shipping for all orders.
2. Packages will be shipped out through DHL, FEDEX, USPS, Royal Mail and International Express Mail Service.
3. Tracking number will be sent after dispatch.
4. Our price does not include any tax fees. Customers are responsible to pay for taxs if there happens to.
5. All products enjoys One Year Warranty.
Order will be shipped to your verified address, please make sure it's correct before paid.