Call: +44 20 3289 8840
Dental Instruments Category
There are two colors, blue and purple to choose, please allow us 2-7 days for product preparation, thanks
5. Display window in Autoclave Sterilizer 16L-II is is larger than old one
6. Pressure unit in Autoclave Sterilizer 16L-II changes to Kpa from Bar
7. In vacuum state, Autoclave Sterilizer 16L-II will show the actual negative pressure while the old one would show “-p”
8. Temperature displays more than one decimal in Autoclave Sterilizer 16L-II, more accurate
9. An increase in the error code helps us judge machine problem
Rated Voltage: 220V±22V, 50Hz±1Hz, 110V±11V, 60Hz±1Hz
Rated Power: 1800VA / 2000VA
Sterilizing (Pressure/temp.): 0.9-1.3kpa/120ºC, 2.0-2.3kpa/120ºC (Both of above are Under Standard Atmospheric pressure)
Fuse-Melting pipe: T15A / T30A
Capacity of water tank: 2L
Water Consumption volume per recycling round: 0.3L
Operating Temp: 0-40ºC
Outside dimensions: 640*530*540mm
1. The autoclave must be plced on the horizontal working base.
2. Must use the distilled water for the purpose of lasting the working life.
3. There shall be no clog to the air cooling window on the outer surface.
4. The object apparatus to be sterilized shall be palced on teh device plate with enough clearnace retained in favor of the air circulation inside the sterilizing room.
5. Empty the water inside the cooling water collecting tank frequently. Usually, the cooling water collecting tank shall be emptied once the water storing tank emptied.
6. Must revolve teh door handle to the teminal side while working.
7. Don't open the door of autoclave when seeing the "0" displayed on teh pressure indicator.
8. Not be too closed to the door of the autoclave in case of scald when open the door.
9. Shut off the power before discharging/installing the sealing ring, and the work shall be carried out after further sufficient cooling in case of scald.
10. Don't drag or drop the autoclave during relocation period of the autoclave.
11. The protecting ground shall be reliable.
12. Equipment must be used away from teh magnetic field.
new display window:
old display window:
1. Orders will be shipped to verified address, please make sure it’s correct before paid.
All shipping address needs PHONE NUMBER.
2. 98% items will be dispatched within 1-3 working days after your payment confirmed.
3. Our paypal account is email@example.com. Money sent is available.
1. Visa, Master Card, Debit are all ok for payment.
2. Credit Card on Paypal is ok.
1. Please contact firstname.lastname@example.org to get more details.
1. Please contact email@example.com to get bank account.
1. ZetaDental.com offers global free shipping for all orders.
2. Packages will be shipped out through DHL, FEDEX, USPS, Royal Mail and International Express Mail Service.
3. Tracking number will be sent after dispatch.
4. Our price does not include any tax fees. Customers are responsible to pay for taxs if there happens to.
5. All products enjoys One Year Warranty.
Order will be shipped to your verified address, please make sure it's correct before paid.