Call: +44 20 3289 8840
Dental Instruments Category
1. The Autoclave is specially designed for the occasions in use of frequent sterilizations. It is equipped with the microprocessor for intelligent control and man-machine interface for convenient operation.
2. With the distinctions in displaying parameter and condition actively, evaluating malfunction automatically, shutting off for self-protection automatically in case of excessive temperature and pressure, this type autoclave is reliable product in sterilizing.
3. This product can be applied in sterilizing for the tooth device, operation device, Sanitary Material and other heat-resisting articles existing in the Dept. of Stomatology, Ophthalmology and lab.
4. Preparation Before Operation
I. Before operation, please connect well to the power source.
II. Connect the power source hole under the back left side of the autoclave and the power socket by the attached power cable inside the product’s package so as to switch on the autoclave.
III. Push down the main power button at the right bottom corner of the front side, when the indicator light is on, it means that the autoclave get through with the shown on the screen.
IV. The autoclave is not heated while under such initialization status.
Rated Voltage: 220V±22V, 50Hz±1Hz, 110V±11V, 60Hz±1Hz
Rated Power: 1800VA / 2000VA
Sterilizing (Pressure/temp.): 0.9-1.3bar/120ºC, 2.0-2.3bar/120ºC (Both of above are Under Standard Atmospheric pressure)
Fuse-Melting pipe: T15A / T30A
Capacity of water tank: 2L
Water Consumption volume per recycling round: 0.3L
Operating Temp: 0-40ºC
Inside dimensions: 200*368mm
Outside dimensions: 650*520*500mm
1. The autoclave must be plced on the horizontal working base.
2. Must use the distilled water for the purpose of lasting the working life.
3. There shall be no clog to the air cooling window on the outer surface.
4. The object apparatus to be sterilized shall be palced on teh device plate with enough clearnace retained in favor of the air circulation inside the sterilizing room.
5. Empty the water inside the cooling water collecting tank frequently. Usually, the cooling water collecting tank shall be emptied once the water storing tank emptied.
6. Must revolve teh door handle to the teminal side while working.
7. Don't open the door of autoclave when seeing the "0" displayed on teh pressure indicator.
8. Not be too closed to the door of the autoclave in case of scald when open the door.
9. Shut off the power before discharging/installing the sealing ring, and the work shall be carried out after further sufficient cooling in case of scald.
10. Don't drag or drop the autoclave during relocation period of the autoclave.
11. The protecting ground shall be reliable.
12. Equipment must be used away from teh magnetic field.
1. Orders will be shipped to verified address, please make sure it’s correct before paid.
All shipping address needs PHONE NUMBER.
2. 98% items will be dispatched within 1-3 working days after your payment confirmed.
3. Our paypal account is email@example.com. Money sent is available.
1. Visa, Master Card, Debit are all ok for payment.
2. Credit Card on Paypal is ok.
1. Please contact firstname.lastname@example.org to get more details.
1. Please contact email@example.com to get bank account.
1. ZetaDental.com offers global free shipping for all orders.
2. Packages will be shipped out through DHL, FEDEX, USPS, Royal Mail and International Express Mail Service.
3. Tracking number will be sent after dispatch.
4. Our price does not include any tax fees. Customers are responsible to pay for taxs if there happens to.
5. All products enjoys One Year Warranty.
Order will be shipped to your verified address, please make sure it's correct before paid.